AW Lake is a corporate from United States of America that sell some product such as Positive Displacement Meters, Turbine Flow Meters, Tricor Coriolis Flow Meter, Lake Variable Area Flow Meter, Lake Paddle Wheel Flow Meter “Flow Stat”, Monitor&Controllers, Flow Sensor&Pickup. One of the product that comes from the categories of Flow Sensor & Pickups is CAPM-2 Carrier Frequency Sensor.
Wiring should be installed by a qualified electrician or instrumentation technician. When dealing with low voltage/power signals from pickups and transmitters, it is important to use a shielded cable between the transmitter and the signal processing unit. A shielded cable will keep most of the electromagnetic interference (EMI) from entering the signal cable and disrupting the signal before it can be processed. A 20-22 gauge 3 or 4 conductor cable with shield is acceptable.
CAPM-2 Carrier Frequency Sensor
The CAPM-2 is a Carrier Frequency sensor, which means there is no permanent magnet in the pickup, and any magnetic drag on the flow meter is avoided. The CAPM-2 requires a supply voltage between 10 VDC and 30 VDC.
A diode in the supply input prevents damage to the module in case the polarity of the supply voltage is wrong. The output signal is a square wave, voltage pulse of approximate amplitude (supply – 1.5 V). The frequency is proportional to the flow rate. For use with JV-CG and JV-KG meters (not for use with aluminum meters).
- Ensure that the flow meter sensor cavity is free of debris prior to installation.
- Screw the CAPM-2 into the flow meter by hand until the sensor nose contacts the bottom of the cavity.
- Do not screw in harder than hand tight, as a thin metal section is under the sensor nose
Meter Compatibility: Used with JV-CG and JV-KG meters (not for use with aluminum meters)
Output Signal: Square wave, voltage pulse of approximate aplitude (supply – 1.5V)
Sourcing Output: PNP type sourcing open collector transistor
Intrinsically Safe: Intrinsically safe CAPM-2o version available
IS Safe Version: UL & cUL certified for use in Class 1, Div 1 locations
Supply Voltage: 10 to 30 VDC
Supply Current: 9 mA @ 15 Volt, Max 18 mA
Minimum Signal: 0.5 Hz
- Square wave
- VHigh ≈ VCC – 1.5V
- VLow ≈ 0V
Duty Cycle: 50%
Frequency Output: Flow dependent up to 2000 Hz
Driving Capacity: 10 mA Max
Temperature Range: -60°F to 185°F (-50°C to 85°C)
Download Data Sheet: CAPM-2 Carrier Frequency Sensor